Dear Sid Ahmed Tadjer.
Mostly biggest part of information about reusable instruments for implantation of artificial articular or metal constructions are written in recommendation of manufacturers. (ISO 17664). In our CSSD we are receive instruments of different manufacturers including motor systems and for all of them we are have information of manufacturers that we can wash them in WD on ordinary validated programs and sterilize them by steam sterilization 134 Degree Celsius- 5 min.
However you should know that all of this special tray, including vendor trays are so heavy(more than 7 kilograms) and you should use longer dry in steam sterilizer or steam pulses of drying to remove humidity from special correctly, and use more strong packaging materials than crepe paper.
Hope it’s help. Petr Demidov, Moscow, Russia